Complete set of nozzles universal for Cold Laser Vityas
- Kit 3 Nozzles
The registration certificate of Ministry of Health RB ¹ to them -
The complete set of nozzles universal (further it is simple
nozzles) represents a set of optical paths from polymer to the device
of quantum therapy "Vityas". The complete set of nozzles allows to
enter the directed radiation directly in bodies and to create optimum
density of a stream of capacity on an output of a nozzle.
|Length nozzles (mm)
|Diameter the bases (mm)
|The connecting size of a radiator (mm)
|Corner at top of a cone (hailstones) *
|Factor of a transparency in a range from 600
nanometers up to 870 nanometers (%) *
* - help sizes;
At carrying out of sessions of laser therapy with optical
nozzles in the field of eyes it is necessary to dress to the patient
Goggles should not have cracks and having chopped off
colour-filters, other defects worsening their protective properties.
Installation and change of optical nozzles should be made
only at the switched off laser radiation of the device.
It is forbidden to direct the radiating ends of nozzles
aside an eye, it is especial at an irradiation of the person.
The operating procedure with nozzles:
- To take the device for the handle and to turn
an aperture of a radiator upwards.
- To take from packing an optical nozzle
necessary for the given procedure and with the help of
the cartridge to fix in a carvingaperture on the part of
a radiator of the device. The cartridge is screwed on a
groove before rigid fixing a nozzle. In order to prevent
damage of a groove to not put significant efforts at
screwing up. If the cartridge and a nozzle have not been
disinfected, preliminary to disinfect them of 70 %
- At use of a nozzle ¹1 in gynecology or
proctology to put on it (her) a condom and to fix it
- To switch on a feed (meal) according to the
passport on the device.
- Preparation for job and choice of an operating
mode is carried out according to methodical
recommendations and the passport on the device.
- At an irradiation of area of eyes to put on
goggles and to the patient.
- To enter a nozzle into a cavity or to
establish the radiating end on the necessary point or a
zone of a body and to keep her(it) in this position
during a session, for example, a hand of the patient.
- To press the button START-UP and to lead
(carry out) a session of laser therapy. If necessary
irradiations of the extended struck sites (a vagina, a
direct gut, etc.) follows in pauses between radiations
slowly and in regular intervals to move a nozzle more
deeply for influence on the following site.
- Upon termination of the established duration
of a session the device automatically will be switched
off with submission of a sound signal.
- If necessary continuations of a session again
to press the button START-UP and to repeat an
irradiation before end of recommended influence.
- After deenergizing radiation to remove (take
- To switch off the device having disconnected
from a network of a feed (meal).
- To remove (take off) a disposable condom.
- To unscrew, disinfect the cartridge and a
nozzle of 70 % spirit and to lay in packing. On it the
session of laser therapy comes to an end.
Practical recommendations on use of nozzles.
To disinfect the cartridge and nozzles follows
rubbing of 70 % by spirit. To process solutions of
formalin, alkalis, etc. active substances it is not
recommended in order to prevent deterioration of optical
characteristics of plexiglas and occurrence of cracks.
Manufacturer's guarantees, service life and
The enterprise - manufacturer guarantees normal
job of nozzles within 12 months from the date of sale at
observance of conditions of transportation, storage and
use, at absence of mechanical damages and thermal
Warranty period of storage of 12 months from
the date of manufacturing nozzles by the enterprise -
At absence in a label of a mark about date of
sale the warranty period of operation is established
from the date of manufacturing nozzles by the enterprise
Storage of nozzles should be made in regular
packing in a room at temperature from +5 up to +40 With
and humidity up to 80 % at any atmospheric pressure.
Complete set of nozzles universal for Cold Laser Vityas.